Glove Box Everyday Routine Maintenance Guide
one) Fill in the use document in time following working the glove box, which include "user, h2o oxygen value, what to do"
two) When changing gasoline cylinders in the glove box, the tension is established to the default price ± 3mbar
three) The everyday doing work force is established in accordance to the working comfort and ease. When not in use for a lengthy time, the tension is established to one-6mbar
4) Often check out the wear of the gloves, test the oil level of the vacuum pump (between the upstream and downstream markings), and increase the pump oil appropriately
five) Thoroughly clean the water probe at the time each two to three months to be certain correct detection. (Also have to have to refer to the precise condition of the setting in various user bins
6) If you generally use natural and organic solvents in the glove box, it is encouraged that you routinely substitute the natural and organic solvent adsorbent material, not later on than one calendar year at the latest
seven) Swap the filter ingredient on a regular basis, no later on than two decades
eight) In accordance to the actual use of your glove box, vac glovebox choose to change the vacuum pump oil cycle, these as black and yellow pump oil, h2o or much more natural and organic solvents in the pump oil. The oil substitute cycle must not exceed 2 decades at the most recent
nine) If a huge amount of organic and natural solvent is inevitably uncovered in the course of the experiment, the cycle requirements to be stopped for the experiment. Right after the experiment, first clean and then commence the cycle
ten) When the setting in the box is harmful or hazardous gas, the exhaust port of the vacuum pump, the exhaust port of the regeneration gas and the exhaust port of the cleaning valve should be exhausted to the outdoor with an exhaust pipe (when exhaust is not authorized to be outside, the exhaust fuel cure need to be accomplished 1st).